Professional, Reliable Medical Products Manufacturer and Exporter.
|Place of Origin:||Jiangxi|
|Brand Name:||EasyThru or OEM|
|Certification:||ISO, CE, FDA|
|Model Number:||1.0#, 1.5#, 2.0#, 2.5#, 3.0#, 4.0#, 5.0#|
|Minimum Order Quantity:||100pcs|
|Packaging Details:||1pc/PE Bag,250pcs/carton|
|Delivery Time:||15-25 days|
|Payment Terms:||L/C, D/A, D/P, T/T, Western Union|
FDA Laryngeal Mask Air Way,
Nontoxic Laryngeal Mask Air Way,
Double Luman LMA Mask Anesthesia
The laryngeal mask can be used to establish artificial ventilation during anesthesia or emergency resuscitation by medical staff in emergency department, ICU and other departments.
2. Parts and Materials
The laryngeal mask is made up of cuff, connector, airway catheter, inflation tube, indicating airbag, catheter joint and charging valve.
The cuff, airway catheter, connector, inflation tube and indicating air bag are made of silicone rubber. Catheter joints and charging valves are made of polypropylene.
3. Key technical indexes
1)The lumen of the laryngeal mask is clean and smooth, with no visible matter to the naked eye; It has no sharp edge, with the parting surface having no obvious dislocation or rough edge; Cuff is colored evenly, with no obvious pattern or color difference; The connector is smoothly connected with the catheter without flash; There is no burr or protrusion in connection between the inflatable tube and the indicating airbag; There is no degumming around the junction surface of the cuff and connector, and the airway catheter is clearly marked.
2)All parts of the laryngeal mask are connected firmly.
3)The inflation tube and valve of the laryngeal mask, the cuff and the indicating airbag have good performance in gas pumping (filling), volume-filling and sealing.
4)All parts of the laryngeal mask are free from blockage and leakage; The mask shall look even with no obvious bulges when the capsule and the indicating airbag are filled.
5)The laryngeal mask should be sterile; The residual amount in the mask sterilized with ethylene oxide should be no more than 10ug/g as leaving factory.
Model:General typeⅠ,General typeⅡ,Silicone single tube typeⅠ,Silicone single tube typeⅡ,Single tube drainage type Ⅰ,Single tube drainage type Ⅱ,Double lumen drainage typeⅠ,Double lumen drainage type Ⅱ, Double lumen multi-tube typeⅠ,Double lumen multi-tube typeⅡ
Specifications for each type: 1.0#,1.5#,2.0#,2.5#,3.0#,4.0#,5.0#
Instructions for use
1. Select the correct size for each patient according to the gender and weight.
2. Head the air valve with an empty syringe to fully deflate the laryngeal mask airway to a very tight vacuum, and moisten the tip back of the laryngeal mask with normal saline to reduce the resistance during the insertion of the laryngeal mask while holding the laryngeal mask in a pen-holding posture.
3. Open the patient's mouth and place the tip back of the laryngeal mask at the back of front teeth; Then pull the tongue with the left hand (or push the tongue with the laryngoscope), and then use the right index finger to assist inserting the laryngeal mask along the back of the hard and soft palate, and continue along the cranial direction until the anterior obstruction is felt in the hypopharyngeal cavity.
4. Fix the catheter of the laryngeal mask with the left hand, use the right hand to inflate the cuff till pressure is shown in the indicating airbag (the filling volume should not exceed the maximum amount indicated ) through the air injection joint with an empty syringe.
Contact Person: Terysa